Application note: cleaning and clean handling
In order to ensure the high performance of our liquid flow sensors in the long term, proper handling and cleaning are essential. Due to the thin flow channel wall of less than 100 micrometers, even the thinnest deposit on the inside of the flow channel can cause measurement deviations. Here are a few basic instructions to follow:
1. Do not let a sensor sit filled with fluid for an extended period of time
2. Do not allow the liquid flow meter to dry out with residual media inside the flow channel
3. For storage, flush with a suitable cleaning agent and dry the flow channel
Check out our application note for detailed assistance on cleaning and clean handling.
Become Part of the Sensirion-World
With Sensirion's preference-based newsletter, we provide you with exactly the information you need. Let us know what content is of interest to you and secure access to exclusive knowledge. Sensirion's newsletters offer insights into current topics, new products, trade fairs and the company as a workplace.